GSI Forum - RDF feed
type: channel
description: GSI Helmholtzzentrum für Schwerionenforschung
items: Anode Wire Problem Summary and Conclusion

Anode Wire Problem Summary and Conclusion
type: item
description: Dear Chamberbuilders, After the discussion last week and with the information from Electron microscope pictures, we have some rolls of Anode wire, which were identified as unusable after the beginning of the roll was identified as 'good' at GSI. Statisitics are not yet really available, so far I know about 2 rolls in Dubna, where the wire is definitely missing gold plating . The REM images sent by Mihai show that there are some little pieces sticking out of the surface under very high magnification, at lower magnification the wire looks fine however, about the other rolls there is no information yet. GSI and Heidelberg has not seen bad anode wire yet (at least not a roll which was good at the beginning and became bad in the middle). The quality of the anode wire that we buy (for everybody) is already the highest quality that Luma provides and that Herbert Stelzer knows of. This means that there is no option to invest money to buy better (or more quality controlled) wire. This means that we have to live with the wire we have, which in turn means that some (additional) quality control step has to be taken: The anode wire of every chamber has to be examined with an electron microscope. The optimal procedure would be to look at the wire just after the winding finished, before gluing the wire onto the chamber. This might cause some delay in the production. I belive that it will not happen very often that the wire goes bad in the middle of a roll, so I think it might be justified to even glue the anode wire onto the chamber, before looking at the wire, if this is necessary. As long as the distance frame is not glued, an exchange of an anode wire grid is possible without too much effort (Oleg and we have some experience in this . There are some indications like gold dust development during winding, which might be an indication for bad wire. However we had gold dust at Heidleberg for 2 rolls of anode wire, which were OK (under REM and under HV and in the test beam). This me…
creator: Clemens Adler
date: 2005-02-11T14:54:51-00:00